See the SDPD Mailing List at : http://www.egroups.com/list/sdpd/

**The 3 "if" :**

- If really the Kalpha-2 component was at an exactly known position relative to Kalpha-1.
- If the alpha1/alpha2 ratio was exactly known.
- If the Kalpha-2 profile shape was exactly the same as the Kalpha-1 profile.

Then we could simulate a Kalpha-1 powder pattern, add to it some
statistical noise, then simulate the Kalpha-2 component, then add the two
patterns and compare...

Mathematically, the two patterns should be equivalent, because of the 3 "if" above, and a Rietveld refinement on both patterns should lead to equivalent results.

Let us try.

The calculation of the 2 powder patterns was done by the Fullprof program on a small-size orthorhombic cell, that of GaOOH :

**Kalpha-1**

See the refinement results in the Fullprof **.sum
file**. Download the **.dat file** and
the **.pcr file**.

**Kalpha-1-2**

See the refinement results in the Fullprof **.sum
file**. Download the **.dat file **and
the **.pcr file**.

**Conclusion**

So what ? Compare the Fullprof .sum files above, do you find any clear difference in the refinements ? There is not. Try a larger cell with thousand reflections will not change anything, if the 3 "if" above are really mathematically/physically true.

Of course, the problem here is that the 3 "if" above are not completely true...

However, this is not a reason for trusting only your eyes, telling that Kalpha-1 is so much better, this would also not be completely the truth.

If the 3 "if" were true, we could perfectly remove the Kalpha-2 component (in case of extremely good statistics, anyway), but we cannot. Nevertheless, we can almost do it, with only some ripple problems, right ? And this is why we prefer to make the Rietveld refinement on the original pattern, but not on the pattern corrected from the Kalpha-2 component : less errors occur when adding the alpha-1 and alpha-2 calculated contributions than when stripping alpha-2 from the observed pattern. Anyway, a primary beam monochromator is nothing else than a Kalpha-2 stripping hardware. Some are not better in that exercise than certain Kalpha-2 stripping software, and may let up to 1.5% of alpha-2 residue on your experimental powder pattern. Of course, if you want your Kalpha-2 stripping sofware to make well the job, you should have a high-resolution pattern, with a counting step sufficiently small, and a quite good counting statistics.

A Kalpha-1 pattern and a Kalpha pattern are almost equivalent. You will not obtain easier indexation from a Kalpha-1 pattern than from a Kalpha pattern corrected from the Kalpha-2 component. And the final Rietveld refinements would be quite similar, in spite of what say your eyes.

Do not believe only your eyes.

Armel Le Bail

October 2000